This Standard sets out the requirements for the following
non-destructive methods for the measurement of the local thickness
of metallic and non-metallic coatings:
(a) Magnetic induction (low frequency) method
The magnetic induction method is used for the thickness testing
of non-magnetic and non-metallic coatings on magnetic
(b) Eddy current (high performance) method
The eddy current method is used for the thickness testing of
(i) a non-conductive coating on a non-magnetic substrate;
(ii) a metallic coating or a non-metallic coating on a substrate
that has appreciably different conductivity.
These methods apply when the coating thickness is not less than
2 μm, and the surface contour permits calibration of the test
The magnetic induction (low frequency) method is suitable for
the measurement of thickness of gold, silver, tin, cadmium, zinc,
copper, chromium, lead and organic coatings, such as paints and
plastics, on magnetic substrates.
The eddy current (high frequency) method is suitable for the
measurement of thickness of oxide coatings on metals. Using
selected frequencies, it is also possible to measure the thickness
of various metallic coatings on either magnetic or non-magnetic
substrates, e.g. tin-lead on copper, silver on nickel/silver,
copper, zinc, cadmium or tin on steel, and copper or silver on a
Under the best conditions of use, these methods are accurate to
±0.7 μm, or ±10%, whichever is the greater.
1 Factors affecting the accuracy of measurement are given in
2 The methods given in this Standard are not suitable for the
measurement of chemical conversion coatings.